YY 0763-2009: Translated English of Chinese Standard. YY0763-2009: Medical endoscopes - Fibre optical cables for lighting
https://www.chinesestandard.net, 2014 M03 17 - 17 pages
This Standard specifies the scope, terms and definitions, requirements, and test methods of fibre optical cables for lighting of the medical endoscopes. This Standard applies to fibre optical cables, which may be used as independent products, used for normal lighting of examination and operation together with medical endoscopes.
What people are saying - Write a review
We haven't found any reviews in the usual places.
according to Equation aperture apply arithmetic sequence bending radius cable being measured cable is placed Cosine radiation panel D65 Standard Illuminant Determination of luminous device of measuring due to falling fibre optical cable horizontally and straightly impact due initial condition instructions integrating sphere Laser light illuminator less than 90 Let the fibre light input surface light intensity light output angle light output surface light transmission performance light transmission rate luminance meter Luminous flux meter luminous flux output manufacturer Measure the luminous measured fibre optical medical endoscope minimum bendable radius moveable flat plate Moveable steel flat nominal value optical cable restore output in accordance performance of initial permissible tolerance placed naturally ranging from 380 relatively spectral radiance sequence of ranging shown in Figure special wooden board specified in sub-clause spectral transmission rate spectrum ranging spherical surface straight condition surface and light terms and definitions Test device torsional value from sub-clause