Implementation of a self-controlling heater
National Aeronautics and Space Administration, 1973 - Science - 13 pages
Temperature control of radiation sensors, targets, and other critical components is a common requirement in modern scientific instruments. Conventional control systems use a heater and a temperature sensor mounted on the body to be controlled. For proportional control, the sensor provides feedback to circuitry which drives the heater with an amount of power proportional to the temperature error. It is impractical or undesirable to mount both a heater and a sensor on certain components such as ultra-small parts or thin filaments. In principle, a variable current through the element is used for heating, and the change in voltage drop due to the element's temperature coefficient is separated and used to monitor or control its own temperature. Since there are no thermal propagation delays between heater and sensor, such control systems are exceptionally stable.
2 pages matching difference amplifier in this book
Results 1-2 of 2
What people are saying - Write a review
We haven't found any reviews in the usual places.
actual temperature ambient range ambient temperature Amplifier A2 analog divider approaches zero breadboarded circuit of Figure control accuracy control heater controlled body controlled temperature Current-sampling delay between heater difference amplifier divider inputs divisor drives the heater electrical Elementary proportional temperature error comparator error comparison FEEDBACK FUNCTION FORWARD GAIN Goddard Space Flight Greenbelt Heat propagation heater and sensor heater current HEATER Figure heater resistance heater temperature heater voltage heater wire heater/sensor loop gain Maxwell G monitor output multiplier offset operational amplifier oscillate output voltage parameters Performing Organization proportional control proportional temperature control reference voltage reference VR Resistance changes resistance sensor resistor Security Classif self-controlling heater self-monitoring heater slope is caused Space Flight Center stable temperature coefficient temperature control point Temperature set reference TEMR test heater thermal propagation delay thermistor transfer function Unclassified varying drive current voltage drop due volts Vy Ro vy values wire filament